I have tried it several times but I have not been able to get it. O ur technicians are military trained PMEL graduates certified with level 9 credentials. What I wanted to do is to get the waveform from the scope. There is a local network and socket is used for A control. It works great but is a bit over my head and I had trouble modifying it to my needs beyond the continuous read. I still cannot figure out why it is not working. Select a valid country.
|Date Added:||25 November 2012|
|File Size:||59.49 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
So far so good For Windows NT, change references to win95 to winnt. How can i solve this problem?
It would be really helpful if any one can explain the process or send some sort of link or text. I’m sure there are moredetails you need – let me know and I will add pieces of code, settings etc.
Keysight Instrument Drivers | Keysight (formerly Agilent’s Electronic Measurement)
We ship “signature required. THe second one shows the configuration zgilent with the error. This is the only way If I can’t do this with other options. See all condition definitions – opens in a new window or tab.
We invite you to join our thousands of satisfied customers. I think I am getting hung up on agulent being sent to the device. I have an arbitrary waveform stored in my A waveform generator. Message 5 of It is not solely a hardware issue, as it only appears while testing certaint variants of our product. We have a couple of spectrum aggilent and lots of signal generators in use.
Message 4 of Skip to main content.
Hope to hear back from you, any advice is greatly appreciated. I am very new so if you can explain the cluster part in a bit easier terms it would be very helpful Domestic customers must also agree to the following terms and conditions: A query interrupted message comes from the machine itself so probably nthing is wrong with the communication but you started a second query before the first one was finished.
Well, my problem is that I am not able to save results in a excel spreadsheet. To achieve this with computer you need to send this command to device. Channel Catalog Subsection Catalog.
Keysight Instrument Drivers
It’s the service after the sale that counts! On A function generator, using RS RSbinary download of arbitrary waveform does not always work. It is because Agilent source make digitized measurements of current and voltage, and after that, the source send to Labview all digitized agilejt together.
I’m an electronics engineering student and interested in instrument control. I got rid of the Zero Switch and changed the current to voltage.
Instrument Control (GPIB, Serial, VISA, IVI) topics
How to force the AWG eg B vii to hold exactly the same last voltage of an arbitrary waveform after having played this ARB waveform? If you go inside the initialize vi or have a serial settings cluster on the frontpanel of the initialize vi fill it in properly and check on the instrument frontpanel if it is reacting.
The first scren capture shows me creating a new IVI session for the A. When I type in a command the response I get is the exact same as the command, and not the actual response.
After this, device sets the frequency to 10 MHZ. I need to know when the waveform starts to be repeated, it is, when the beginning of the waveform oc Message 1 of